We know there are plenty of differences when it comes to used equipment and quite often, choosing between different pieces is difficult, especially when the equipment is not sitting right in front of you. Commands to control the same class of instrument, e. Not just a picture from the manufacturer’s website, but the actual piece of equipment you would receive. Help improve your experience on DHgate. Manuals, E, Drivers, Links. Although the electrical and physical interfaces were well defined, there was not an initial standard command set.
|Date Added:||3 February 2004|
|File Size:||42.74 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
You can zoom in close to see the labels with the serial number or zoom out to see the overall condition of the equipment. Articles needing additional references from February All articles needing additional references Commons category link is on Wikidata Wikipedia articles with GND identifiers.
Devices from different manufacturers might use different commands for the same function. The multi-conductor parallel data connectors and shielded cable were 8230a more costly than the connectors and cabling that could be used with serial data transfer standards such as RSRSUSB, FireWire or Ethernet.
Valid Email Required Item to Inspect: We have longer warranties than our competitors. To solve the problem,you a contact rongtong.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
In Stock Call for Quote. The slowest participating device determines the speed of the bus. Trade-in service or analytical equipment: The standard allows up to 15 devices to share a single physical bus of up to 20 meters total cable length.
Well, what if you were able to see a piece of equipment before you purchased it? Early versions of the standard suggested that metric screws should be blackened to avoid confusion with the incompatible UTS threads.
Hl to possible delay of exchange rate update, price in various currencies is for reference only. Systems 26 Autosamplers 13 Pumps 12 Detectors One of our friendly Customer Service representatives will be contacting you shortly with information on when the equipment you requested will be available for viewing.
Keysight Agilent HP E a PCI Hp-ib HPIB GPIB Interface IEEE | eBay
P P P P Np Please contact one of our Customer Support Specialists if you have any questions about what is included with this equipment or if you require any additional information. Please note that any additional items included with this equipment such as accessories, manuals, cables, calibration data, software, etc.
They are sometimes called “Centronics connectors” after the pin micro ribbon connector Centronics used for their printers.
From Wikipedia, the free encyclopedia. In the late s, Hewlett-Packard HP  manufactured various automated test and measurement instruments, such as digital multimeters and logic analyzers.
Keysight Agilent HP E2078 82350a PCI Hp-ib HPIB GPIB 488 Interface IEEE 3188.012
February Learn how and when to remove this template message. We know there are plenty of differences when it comes to used equipment and quite often, choosing between different pieces is difficult, especially when the equipment is not sitting right in front of you. We will match prices with our competitors. Click a thumbnail to view larger image: Thanks for looking and we look forward to hearing from you! We have sufficient export experience, including professional packing and proper documentation.
Although the electrical and physical interfaces were well defined, there was not an initial standard command set. Every device on the bus has a unique 5-bit up addressin the range from 0 to 30 31 total possible addresses. Manuals, Datasheets, Drivers, Links.
They are held in place by screws, either UTS now largely obsolete or metric M3. It has been tested at SpectraLab Scientific. Please 28350a the box if you would like to receive future offers or Newsletter. There was no requirement for galvanic isolation between the bus and devices, which created the possibility of ground loops causing extra noise and loss of data.